Title & Document Type: A 50 MHz Programmable Pulse Generator This manual may contain references to HP or Hewlett-Packard. Hp A Pdf User Manuals. View online or download Hp A Operating, Programming And Servicing Manual. View and Download HP A operating, programming and servicing manual online. 50 MHz programmable. A Pulse Generator pdf manual download.
|Published (Last):||28 January 2017|
|PDF File Size:||2.88 Mb|
|ePub File Size:||17.4 Mb|
|Price:||Free* [*Free Regsitration Required]|
Fast and Easy Operation To make operation easy, GPH interface 8112q the operator in a straightforward manner by which the front panel display always shows the parameter being varied and its value. The timer is conflgured as an astable multivibrator which produ ces an output of approximately Hz when enabled when TIRE is active low.
8112A 50 MHz Pulse Generator
Check the Width Generator operation against Table Our service centers may be able to perform calibration if no repair parts are needed, but no other support from Agilent is available. Orient the fan so that the arrow on its case indicating the direction of air-flow: Allocate memory for max- imum reply length Request current settings of instrument Read reply into allocated memory Print the reply.
Report item – opens in a new window or tab. Slope Error Bit 3 This error is caused by excessive slope. Interrogate Error lERR ! For this purpose the instrument contains the Status Byte register: Manual Trigger This key can be used to simulate the external trigger signal.
HP A OPERATING, PROGRAMMING AND SERVICING MANUAL Pdf Download.
The HP A is a fully programmable 50 MHz pulse generator with 5 manial transitions and 32 Vpp into open circuit max output manjal. Full text of ” HP: Switch on the HP A and check that error number E52 is displayed. Note You may wish to set up Output Limits as described in Chapter 4 Operating to protect the device under test. This manual does not discuss the HP-IB protocol or hardware.
The range of delay between the two pulses is 20 ns to ms. Make a copy in order to record your test results. Repeat last two steps and re-adjust if necessary. Errors read with lERR: Refer to Chapters Deletion of W3 B 81112a It is possible to increase h decrease rise and fall times of output pulses.
Blocking Flip-flop The blocking flip-flop UA performs two functions: Item description Seq which among other things prohibit About half the interior pages are separated from the binder, but does effect readability.
hhp The self-prompting operation and HP-IB programmability of the HP A ensure that it is manuwl and easy to use in stand-alone and automatic-test applications.
Back to home page. Switch off the HP A 3. A continuous pulse train is generated. Instrument Overview and Troubleshooting Guide For warranty service or repair, this product must be returned to a service facility designated by HP. The front panel controls are disabled apart from: Limited output mode is switched off by pressing the limit!
The oscilloscope response is very slow for measurements with PER in the ms range. Adjust the range high end with A2R3 for Check the Delay Generator operation against Table Correct Delay Control 4.
HP 8112A Operating, Programming And Servicing Manual
The foregoing warranty shall not apply to defects resulting janual improper or inadequate maintenance by the Buyer, Buyer-supplied software or interfacing, unauthorized modification or misuse, operation outside of the environmental specifications for the product, or improper site preparation or maintenance.
If re-adjustment is necessary, repeat Fixed, Linear and Gauss adjustments as required. Check that the correct line fuse is fitted. Period Control Verification Test Examine the HP A manuall on the scope display in order to verify that the pulse characteristics do not exceed the specified limits as identified in Figure Name of this program: The system controller can therefore monitor this flag to detect when a programming message has been implemented.
Period Generator ramp signai pin 20 3. Operation of the burst control circuit can be categorised as follows: See Details – opens in a new window or tab.
This amplifier compares the virtual ground at the inverting-input of the main-amplifier with actual ground, and supplies a compensating current via Rc in order to maintain zero difference. Verify that the disable 1 key disables the output signal.
Address Decoders The address decoders can be checked using signature maunal Adjust the range low end with A2R57 for Address 0 – 30 front panel selected General Memory Non-volatile stores up to 99 complete panel settings.